New Arrival X-ray Machine Fluoroscopy Portable digital medical x ray machine Price
Output Image size
X-ray tube voltage
X-ray tube voltage deviation
Tube current deviation
Generator working frequency
Stand-Position Protection Area
operation method: 1, The use of the first check the appearance of the box in the transport process for damage. Out of the box to check whether the host has a collision damage traces. 2, Then remove the host, the additional protective screen at the front of the three gap at the front of the three screws into the rotation angle of an additional protective screen can be fixed. 3, Turn on the power of the switch (insert the power plug into the three-hole single-phase power socket with grounding wire). The switch power indicator light is on, and the digital control panel lights. 4. Press the start button, the digital display panel shows the anode voltage (KV) or the anode current (μA) of the X-ray tube (depending on the selection switch), press the switch to display the anode current, press the switch again Shows the anode voltage, indicating the instrument has been working, this time the screen was yellow-green. 5, The object is placed between the decorative tube and the liquid crystal display, (objects as close to the liquid crystal display), you can see the measured object in the fluoroscopy perspective image. (KV) adjustable knob, the voltage knob clockwise rotation, the current value is small, the current value is small; the voltage (KV) adjustable knob, the voltage knob rotation clockwise, The voltage value becomes larger; counterclockwise rotation; voltage value becomes smaller, until the image clarity is satisfactory. 7, In ensuring the premise of a clear image, should be possible to use low current (μA) for perspective. 8, When the hand presses the start button, the X-ray tube sends out the ray, has the work indicating lamp instruction on the digital display operation plate, presses the start button again or the fixed time time arrives, the X ray tube stops working (timing set in 1 minute about). 9, The machine should be intermittent use, if the continuous working time is too long, more than the machine effective operating temperature range, the machine will automatically shut down, at this time on the digital display panel over-temperature indicator light, need to stop a moment, until the temperature decreases Recovery time, depending on the ambient temperature. 10, Fluoroscopy as far as possible so that patients stay away from the ray focus and close to the LCD side, reducing the absorption of human skin dose.
working principle: When the power is turned on and the start button is pressed, the unit starts operation. The first pulse signal from the main circuit board, the power amplifier, voltage doubler circuit generates high voltage to the X-ray tube anode; at the same time the main circuit board issued by the second pulse signal amplified to the X-ray tube filament to provide power to X Ray tube produces X-ray, and through the digital display panel shows the corresponding value (KV / μA). The object to be measured is placed between the exterior tube (containing the X-ray source) and the liquid crystal display, and the liquid crystal display screen shows a sharp image of the object to be seen. In order to make the X-ray machine work stably and reliably, the system adopts pulse-width modulation technology to keep the tube current and tube voltage constant and the X-ray tube works in the best condition. And a high-pressure slow start function, so that no high-pressure X-ray tube anode overshoot phenomenon. The main controller uses micro-chip devices, and 20KHz frequency work, greatly improving the efficiency of the entire system, eliminating the audible noise, for the operator to provide a quiet use of the environment, but also reduce the size. Portable X-ray machine using high-frequency high-efficiency switching power supply, itself has overcurrent, overvoltage protection. To ensure the safety of portable X-ray machine, machine with a variety of protection devices, making it safe and reliable work.
Note: General precautions: Please read the instruction manual carefully before use. 1, check the switch power supply connection is correct before use; check whether the machine is normal, boot normally before use. 2, due to portable X-ray machine thermal capacity subject to certain restrictions, the proposed operation should pay attention to intermittent work. 3, the product factory with precision instruments for a comprehensive proofreading, the user is strictly prohibited to open their own adjustment. 4, do not use alcohol or acidic reagents and other organic solvents to wipe the surface, to avoid damage caused by corrosion and moisture. If it is necessary to clean, wring clean towel wipe, and to be dry and then use the machine. 5, shielding protection: This product is X-ray diagnostic equipment, radiation exposure during the machine, the user should be in the rear area of the image to operate, you can wear effective protective equipment. 6, time protection: all staff to minimize the stay in the X-ray field time. 7, distance protection: all unrelated personnel try to stay away from the X-ray source.
Note on Electromagnetic Compatibility: Electromagnetic compatibility testing claims the basic performance: machine work, the LCD display brightness is normal, the machine power supply is normal. 1, The product should not be used in strong electromagnetic interference environment; 2, In the use of this product should not have electromagnetic interference with the device at the same time use; 3, Portable X-ray machine with 24V switching power supply output lead length of 1.2m, does not allow private lengthened. 4, The user should not replace the whole machine parts, so as not to change the device anti-jamming. 5, Portable X-ray machine should not be stored with other electromagnetic interference devices and close to and superimposed.
1. For all your inquires about us or our products, we will reply you in detail within 24 hours. 2. We own professional team with professional attitude to recommend products and services, introducing the products for you. 3. Our products have various type and function, We will recommend to you that according to your requirements ; The products have one year free warranty, after which they can still be provided after-sales service. 4. We offer OEM services, And we can print your own logo on the products. 5. We have well-experienced engineers, who can help you better use our products.
Your satisfaction is our duty! It will be our great honor to serve you. We do hope we can offer you more help in medical field and build the long term business relationship with you.
Q: How to buy your products ?
A: We have some products in stock , you can take away the products after you arrange the payment ; If we don’t have the products in stock you want, we will start production once getting the payment.
Q: What’s the warranty for the products ?
A: The free warranty is one year from the date of commissioning qualified.
Q: Can we visit your factory ?
A: Of course, welcome to visit our factory if you come to China.
Q: How long is the validity of quotation ?
A: Generally, our price is valid within one month from the date of quotation. The price will be adjusted appropriately as the price fluctuation of raw material in the market.
Q: What’s the production date after we confirm the order ?
A: This depends on the quantity. Normally, for the mass production, we need about one week to finish the production.